Achieving durable local control (LC) for larger (e.g., >2-3 cm) brain metastasis whether newly diagnosed or recurrent remains problematic. Resection (R) alone is typically insufficient and adding radiation therapy (RT) still results in a 12-month recurrence rate of 20% or more in many series. Hypothesizing that R plus immediate radiation utilizing brachytherapy may improve outcomes for this cohort of patients, we designed and prospectively evaluated a permanently implanted surgically targeted radiation therapy (STaRT) device consisting of cesium-131 (Cs-131) seeds positioned within a collagen carrier (GammaTile, GT Medical Technologies, Tempe, AZ). The device was designed to prevent direct source-to-brain contact and maintain inter-source spacing after closure.
This was a subgroup analysis of a cohort of patients with either recurrent or previously untreated brain metastases enrolled in a prospective, multi-histology single-arm trial (ClinicalTrials.gov, NCT#03088579), conducted between February 2013 and February 2018, of resection and tumor bed brachytherapy with Cs-131 containing permanently implanted collagen tiles to deliver 60 Gray (Gy) at .5 cm depth. No additional local therapy was given without progression.
A total of 16 metastases in 11 patients were treated; 12 tumors were recurrent and four were previously untreated. The median preoperative maximum diameter was 3.2 cm (range: 1.9-5.1 cm). Histology was seven breasts, six lungs, and three sarcomas. The median age was 60 years (range: 41-80 years); the Karnofsky Performance Status (KPS) was 70 (range: 70-90). The cohort consisted of seven females and four males. The mean time for implantation completion was five minutes. The median overall survival (OS) was 9.3 months. At a median radiographic follow-up of 9.5 months' treatment, site progression was found in 1/16 (6%) at 10.9 months, and the median treatment site time-to-progression (TTP) has not been reached [95% confidence interval (CI): >10.9 months]. At 12 months, the Kaplan-Meier (K-M) estimates for LC after R+STaRT for all tumors was 83%; for previously untreated tumors, LC at 12 months was 100% and for recurrent tumors, it was 80%. Two tumor beds (12.5%) experienced radiation brain changes: one had grade two and the other grade three. No surgical adverse events occurred.
In this single-arm precommercial study, R+STaRT demonstrated excellent safety and LC in this cohort. The device has recently received FDA clearance for use in newly diagnosed and recurrent brain metastasis, and randomized clinical trials vs. standard of care treatments in both settings are scheduled to open in 2020.
It is estimated that 20-40% of all patients with cancer develop brain metastases and that 200,000-300,000 new brain metastases are diagnosed in the United States each year [1,2]. The size of the metastatic lesion often drives the therapeutic approach, and smaller, asymptomatic lesions in patients with known malignancies are frequently treated using radiation therapy (RT) alone and larger (e.g., >2-3 cm), symptomatic, and/or recurrent lesions in accessible locations are frequently resected [2,3]. With routinely utilized modalities, the 12-month surgical bed local control (LC) rate for smaller brain metastasis is about 90%; however, for larger lesions, the 12-month recurrence rate is 40-60% with surgery alone and reportedly 20% or more even with adjuvant RT [4-6]. This dichotomy in treatment outcomes based on size was highlighted in a recent randomized trial . As patients with brain metastasis may now experience longer life expectancies owing to advances in systemic and targeted treatments, the impact of this relatively low LC rate for larger lesions means that many patients will need additional brain-tumor-directed procedures within one year of their initial treatment, with each additional procedure contributing to the potential for toxicity, diminished quality of life, and more expenditure [2,3,7].
There is a general consensus that RT is the single most effective postoperative treatment to help prevent recurrence of brain metastasis, but there is currently no consensus as to the “best” type of RT treatment in the adjuvant setting. Several topics of active debate exist regarding postoperative RT brain metastasis care, including toxicity from whole-brain coverage, optimal radiosurgery (RS) fractionation (single vs. multi-fraction), RS treatment margins, ways to deal with postoperative cavity dynamics, and the frequent exposure of external beam radiation therapy (EBRT)-transited tissues to near-tolerance doses of radiation [5-12]. These issues are all currently unresolved and likely contribute to the variable control and adverse event outcomes reported for similar tumor types and sizes [2-9].
To optimize the dosimetry and tumoricidal aspects of postoperative adjuvant radiation, a number of groups over the years have implanted radioactive sources under direct visualization at the time of the craniotomy, a technique referred to as brachytherapy [13-15]. Brachytherapy can be especially useful in the focal treatment of larger tumor beds where single-fraction stereotactic radiosurgery (SRS) or fractionated stereotactic radiosurgery (FSRS) may expose larger volumes of "innocent bystander" tissue resulting in the need to reduce the RT dose and/or the coverage area, either of which can potentially limit the effectiveness of RT in initial therapy [2,5,11,12] and recurrent disease [3,16].
A relatively new development in brachytherapy has been the commercial availability in “seed” form of the short half-life (t ½) gamma-emitting isotope cesium-131 (Cs-131) . This isotope has a similar energy, and therefore treatment depth, as iodine-125 (I-125), the most commonly used radioactive isotope in intracranial and extracranial tumors. The major difference between these isotopes is that Cs-131 has a t ½ of 9.7 days compared to 59.4 days for I-125, and this markedly shorter half-life allows a much more rapid dose delivery, diminished time of radiation exposure, and possibly more rapid tumor control [18,19]. Published studies about Cs-131 to date have used a variety of doses and implantation techniques, and outcomes appear to match or exceed what is currently achievable with EBRT approaches [20-23].
Given the above-mentioned background, we hypothesized that combining maximum safe resection with the immediate initiation of radiation could improve patient outcomes in larger brain metastases. To mitigate some of the shortcomings of prior brachytherapy approaches, we designed a biocompatible device utilizing Cs-131 that prevents direct source-to-brain contact and maintains inter-source spacing after closure. This is the first prospective report about the use of this surgically targeted radiation therapy (STaRT) device in brain metastasis.
Materials & Methods
Trial design and participants
This was a subgroup analysis of the brain metastasis cohort associated with a larger prospective, single-arm, multi-histology single-institution study using a prototype medical device (ClinicalTrials.gov, NCT#03088579). The larger trial involved 108 brachytherapy procedures in 96 adult patients with aggressive intracranial neoplasms. The institutional review board (IRB)-approved informed consent was obtained from all participants. Major entry criteria included newly diagnosed or recurrent neoplasm; a planned resection; determination by the treating physicians that resection alone was unlikely to be sufficient to prevent further local recurrence; total prior same-site radiation dose of <100 Gy; capable of at least self-care (KPS of 70); and life expectancy of >six months; the trial allowed uncontrolled disease outside the operative field. Given the anticipated enrollee variability, we specified a uniform treatment of a standardized radiation prescription and implantation technique after maximum safe surgical resection. Initial results from the larger trial, the outcomes in 20 consecutive recurrent previously irradiated meningiomas, have recently been reported .
The technique has been previously described in detail . To summarize it briefly, the expected postoperative resection bed surface in cm2 was estimated from a preoperative MRI, and the number of seeds required to construct the needed collagen carrier(s) with the ratio of one seed/cm2 of surface area was ordered in advance. The prescribed dose for all patients was 60 Gray (Gy) at a depth of .5 cm from the operative bed surface, and the design of the device in terms of seed activity and source placement within the carrier was optimized to facilitate this. During the resection in the operating room, and using a sterilized, shielded reusable stainless steel loader (GammaTile Loader, GT Medical Technologies, Tempe, AZ), the Cs-131 seeds in suture (Proxcelan, IsoRay Medical, Inc., Richmond, WA) were precisely embedded into lyophilized collagen squares (Suturable Duragen, Integra LifeSciences, Plainsboro, NJ). The Cs-containing collagen squares (referred to as “tiles” when assembled) were constructed with the radioactive sources embedded equally and symmetrically spaced at 1.0-cm intervals when considered from above or below, but with asymmetric spacing in terms of the source depth with a .3-cm offset from the bumpy side of the tile surface vs. .1 cm from the smooth side (Figure 1).
Craniotomy and maximal safe resection were performed in the usual fashion. If frozen section pathology did not confirm neoplasm, implantation was not performed. After completing resection and assuring hemostasis, the tumor bed was lined with the Cs-131-embedded collagen tiles. Tiles were typically placed against the walls of the resection bed oriented with “bumps to brain” (i.e., .3 cm from the radiation source to tile surface) to achieve the desired dosimetry. Tiles tend to remain where placed due to the hydrophilic nature of the collagen, with fibrin glue or other adhesives utilized at the surgeon’s discretion. Wound closure was accomplished in the standard fashion, with the replacement of native cranium in all cases. The operating room was surveyed throughout the procedure to ensure staff radiation exposure was in compliance with applicable statutes .
All patients received routine postoperative care and written discharge instructions appropriate for the surgical procedure and radioactivity levels present at the time of hospital discharge . No additional local therapy was given without progression.
Postoperative MRI and stereotactic, noncontrast, thin-cut CT scans were obtained within three days for all patients, and the extent of surgical resection (subtotal resection vs. gross total resection) was determined from these exams. Post-implant dosimetry was calculated using commercially available treatment planning software (BrachyVision, Varian, Palo Alto, CA). An example case is shown in Figure 2.
Follow-up visits and MRIs varied according to clinical needs but were typically scheduled every three months for the first year and every four to six months thereafter. Local failure was defined as new or progressive enhancement within 10 mm of the operative cavity at any time during follow-up, a positive biopsy, or the delivery of additional same-site local therapy.
Clinical outcome measures
Outcomes investigated included treatment site LC (including overall LC by size and by prior treatment), treatment site time-to-progression (TTP), treatment site TTP after the most recent prior same-site treatment (for previously treated tumors), overall survival (OS), distant brain failures (DBF), number and type of subsequent therapies, and radiation and surgical adverse events using Common Terminology Criterion for Adverse Events (CTCAE) Version 5.0. .
TTP was determined using the standard Cox proportional-hazards model and was used to assess the effect of variables on TTP. Survival estimates of TTP were given, with 95% confidence intervals, where appropriate. We also used lognormal frailty models to control for the effect of the individual patient and individual tumor on TTP . Analyses were conducted using SAS 9.4 (SAS Institute Inc., Cary, NC).
The number of months was calculated as the number of days divided by 30.4. For analyses at the site level, standard errors and confidence intervals were not adjusted for multiple sites per subject. We had one subject with four sites, one subject with three sites, and nine subjects with one site each. OS was calculated based on the date of the initial implant.
Patient demographics and characteristics
From February 2013 to February 2018, 16 separate metastatic lesions in 11 patients were treated on the trial. The median maximum preoperative diameter was 3.2 cm (range 1.9-5.1 cm). At the time of enrollment, 12 treatment sites (75%) were recurrent after prior treatment, and four sites (25%) were previously untreated. For previously treated patients, prior same-site resections were none in three patients, one in four patients, two in three patients, and three in two patients (Table 2). Prior same-site RT courses were none in one patient, one in five patients, and two in six patients; the median prior same-site RT dose was 32 Gy (range: 15-110 Gy) (individual dose and modality details are provided in Table 2). The median age at implant was 60 years, and the median KPS was 70 (range 70-90). The cohort consisted of seven females and four males. Tumor types treated were breast (seven), non-small cell lung (five), sarcoma (three), and small cell lung cancer (one). Lesion location was recorded as parietal in six, frontal in four, temporal in three, posterior fossa in two, and parietal-occipital in one. Gross total resection was accomplished in all patients (Table 2).
The median observation period for all patients was 9.3 months (range: 1.4-28 months), and median radiographic follow-up after tile placement was 9.5 months (range: <1-25.2 months) (Table 1). Two patients expired <90 days from surgery, one from cardiac arrest and the other after pathological hip fracture and subsequent hospice referral (Table 2; cases 5 and 11, respectively). Excluding these early deaths, the median OS was 12.0 months (range: 3.3-28 months) (Table 2). No patient was lost to follow-up.
The mean number of seed sources used was 19 (range: 5-63), and the mean activity implanted was 71 mCi (range: 18.5-227 mCi) (Table 1). The added time to surgical resection for tile placement ranged from two to 22 minutes (median: seven minutes) and dropped during the course of the trial to under five minutes.
By maximum preoperative diameter, the Kaplan-Meier (K-M) estimated LC at 12 months for all tumors, tumors of <2.5 cm, and those of >2.5 cm was 83%, 100%, and 75%, respectively (Figure 3A). Treatment site progression occurred within one implanted area (1/16, 6%) at 10.9 months (Figure 4). Following treatment, the median treatment site TTP for the group has not been reached (95% CI gives a lower limit of at least 10.9 months).
For tumors with no prior same-site treatments, no treatment site failures were encountered, and the K-M estimated LC at one year was 100% (Figure 3B). For tumors with previous same-site treatments, 1/12 (8%) failed at the treatment site, and the estimated LC at one year was 80% (Figure 3B).
Time to treatment site progression after R+STaRT vs. time to treatment site progression after most recent prior treatment: for tumors with previous same-site treatments (n = 12), we examined the treatment site TTP for each tumor after trial therapy and compared this to treatment site TTP after the most recent prior treatment. The median treatment site TTP for the prior treatment was 4.8 months (95% CI: 1.9-22.0 months) and the median time after R+STaRT has not been reached (95% CI gives a lower limit of at least 10.9 months). Treatment site LC at 12 months with the prior treatment was 33% vs. 80% after R+STaRT (SE: 13.61 and 17.89, respectively). With site-level frailty term, hazard ratio (HR) was 0.052 (p = 0.0073; 95% CI: 0.006-0.452). With subject-level frailty term, HR was 0.061 (p = 0.0095; 95% CI: 0.007-0.504). With no frailty term, HR was 0.082 (p = 0.0175; 95% CI: 0.010-0.645). In single-variable models, age, gender, number of prior surgeries to index site, number of prior SRS events to index site, and number of prior systemic to index site were not significant at the 0.05 significance level.
At a median observation period of 9.3 months (range: 1.4-28 months), five patients (45%) remained alive. The median survival was estimated at 9.3 months (95% CI: 1.7-21.3 months). OS was calculated on a per-patient basis, based on the date of the initial implant.
No patient remained hospitalized beyond what was typical for the surgical procedure. Two patients expired <90 days, one from cardiac arrest and one after a pathological hip fracture (Table 2; cases 5 and 11). Two implant sites (2/16, 12%) exhibited radiation brain injury (RBI) after implantation and fully resolved with two- and four-week courses of dexamethasone, respectively (Table 2, cases 3a and 8). Both were in patients who had received same-site RT prior to the implantation. No deaths were related to the reoccurrence of a treated lesion (Table 2), and no staff or caregiver toxicities occurred.
Subsequent brain therapies
After the implantation, six patients experienced intracranial failure. Five patients experienced only DBF and all underwent additional brain treatments, with whole-brain radiotherapy (WBRT) used in one patient and four patients receiving a total of nine site-specific treatments: SRS in five cases, on-trial brachytherapy in three, and laser interstitial thermal therapy (LITT) in one (Table 2). The one patient who experienced both DBF and implant bed failure underwent salvage SRS for both (Table 2, Figure 4).
All radiation exposure readings during implant construction, operating room use, and discharge were at acceptable levels per statute . At the end of the cranial closure, all patients had exposure levels below 6 mR/hr, the typical level specified for home discharge . Upon discharge, patients were given instructions related to their recorded radiation level and living situation, to minimize radiation exposure to family members/caregivers.
In this report, we presented our experience with surgery plus a prototype implantable radiation device in the treatment of patients with larger recurrent or newly diagnosed brain metastasis enrolled in a prospective single-arm observational trial. The implant utilized was a novel biocompatible device that holds multiple radiation sources while also functioning as a three-dimensional source spacer.
The path to an optimum outcome for patients with larger metastasis is not well defined either in the initial or recurrent setting. The current National Comprehensive Cancer Network (NCCN) guidelines note that surgery is “preferred” for newly diagnosed limited disease patients, and include resection as an option in most other scenarios, including for the setting of recurrent brain metastasis in patients with “limited” or “extensive” disease . These guidelines also include references to repeat irradiation if “further RT possible,” as repeat RT use is often limited by toxicity concerns, not lack of efficacy .
Some of the known shortcomings of adjuvant EBRT in either initial or recurrent brain metastasis settings include 1) an increasing injury risk as the irradiated brain volume increases; 2) tumor regrowth during the time needed for initial wound-healing prior to starting EBRT; and 3) a difficulty on postoperative imaging in precisely determining the area(s) needing treatment [9-12,16]. While post-resection brachytherapy with low-energy sources can potentially treat a smaller volume of normal tissue than EBRT (including SRS) and also allow the initiation of treatment immediately, it is not without drawbacks and has not achieved routine use status in brain tumors [13,27]. Previously published brachytherapy series have pointed out several problems with existing implant techniques and devices including a) the potential for tissue injury from high radiation doses arising from direct tissue contact with either surface placement or intra-parenchymal insertion of sources; b) determining and maintaining the precise inter-seed operative bed geometry needed for accurate dose delivery; and c) surgical team and caregiver concerns regarding radiation exposure [13-15,18,20-22].
LC in brain metastasis is often benchmarked to one year. Our LC of 83% at one year for this group of very challenging patients was encouraging (Figures 2, 3), particularly as achieving freedom from local progression is more difficult when treating larger and/or recurrent disease [3-6,8,9,16]. Our one-year LC was similar to the findings from Brown et al. of 80% for resection and adjuvant WBRT in previously untreated patients . While the use of adjuvant WBRT eliminates concerns over precisely where to treat, this tradeoff is increasingly unpopular clinically [2,6,7]. We and other centers investigating brain brachytherapy for metastasis believe that it may be particularly useful in larger lesions as control appears relatively independent of size [20-23]; our study results support this finding. Figure 2 shows our LC by size grouping and with only one failure. Interestingly, this failure was in a patient with a 3.4-cm sarcoma metastasis who developed four new intracranial lesions >10 months post-implant. At the time of SRS to these new lesions, a small area of persistent enhancement in the implant area was also treated with SRS, and this was scored as a failure; he died of systemic spread at 23 months (Figure 4; Table 2, case 1a).
Unlike EBRT therapies, brachytherapy allows adjuvant treatment to start immediately. To take full advantage of the minimum tumor burden, we chose to use the 9.7-day t ½ Cs-131 as the shorter t ½ of Cs-131 delivers 50% of the therapeutic dose within <10 days of surgery and 88% of the dose within 30 days vs. 60 days for 50% delivery and 300 days for 88% dose with the more commonly used isotope I-125 . This markedly shorter time for dose delivery has been postulated to offer a significant advantage in the treatment of tumors exhibiting a relatively short doubling time .
Historically, brain brachytherapy literature has been associated with high rates of brain necrosis and/or wound complications . Despite the heavily pretreated nature of many of our patients, the adverse event rate was at or below what was expected. No wound complications related to the implant procedure were seen (Table 2). Two of the 16 treatment areas developed evidence of radiation brain changes, both at sites that had previously undergone SRS, and neither patient required long-term steroids (Table 2). At similar follow-up times, findings in published literature related to traditional brachytherapy for radiation brain injury complications range as high as 24-60%; SRS in re-irradiated patients report 11-16% necrosis rates, and surgical complications in neoplasm resections have been reported to be between 1-24% for dural closure-related complications and .5-4% for wound infections, respectively [3,12,13,16,20,22,28]. One explanation for the low brain injury rate observed in our experience vs. prior brachytherapy publications may be attributed to the difference in implant technique. Prior brachytherapy series have often positioned the seed sources directly on or within the substance of the brain, resulting in extremely high tissue doses at the point of contact, potentially exceeding tolerance [13-15,22]. In contrast, the collagen device used in this trial, when whetted by cerebrospinal fluid, was specifically designed to offset the sources 3 mm from the brain. Even though this is a relatively small distance, by leveraging the physics principle known as the inverse square law, the design of the device lowers the resection bed surface dose to ~120-150 Gy, falling rapidly to ~60 Gy at a depth of .5 cm . Additionally, the use of the collagen tiles provided fixed inter-source spacing during placement and minimized post-closure source shifting, lowering the risk of areas of localized overdose (“hot spots”) from source clumping or areas of underdose (“cold spots”) that could lead to tumor recurrence . We believe these aspects of the tile design allowed a more predictable and uniform radiation dose than had been previously achievable in brain brachytherapy and that this design may have contributed to the lessening of necrosis and wound breakdown. Also, in previously irradiated tumors, confirming recurrence with intraoperative pathology prior to implantation is part of the workflow (see Materials and Methods). Thus, the likelihood of adding additional radiation to symptomatic radiation injury is almost eliminated. One additional possibility is the treatment prescription dose, 60 Gy at .5 cm, which was relatively “conservative” in comparison to other brain brachytherapy series [13-15,20,21]. The specification of this dose was a deliberate attempt to lessen the necrosis risk given our heavily pretreated patient cohort. Nevertheless, this dose seems to have been sufficient for tumor control (Figure 3).
Potential study limitations
Our data were drawn from a subset cohort that was part of a larger, single-arm, prospective observational trial. The smaller number of patients involved, the lack of a control group, and the single-site nature of the trial present several challenges for data interpretation. This proof of concept trial was deliberately designed to accrue quickly by enrolling patients with varied metastatic histologies, controlled and uncontrolled disease outside of the operative field, and varied prior treatments. As reported in detail in Table 2, most patients had recurrent disease and had been heavily pretreated.
Despite this being an inherently unfavorable cohort, the median OS for all patients was 9.3 months, and we believe this survival was probably sufficient to judge the harms and LC benefits. For assessment of surgical harms, typically judged within 90 days, the reported follow-up appears sufficient . In terms of radiation brain changes, the answer is also that it was probably sufficient . A recent meta-analysis of harms in brain brachytherapy re-irradiation used a minimum of six months of follow-up post-second treatment for study inclusion, and our median follow-up was 9.5 months. LC assessment was also likely long enough in that that the median TTP after R+STaRT has not been reached vs. the median treatment site TTP for the prior received treatments of 4.8 months (Table 2).
One question that arises is whether the durable LC was due to surgery alone. For any one case, this is a possibility, but we feel this is generally unlikely for multiple reasons. First, the majority of enrolled patients had progressed despite both prior same-site surgery and prior adjuvant radiotherapy (Table 1). Secondly, as almost all patients received their prior care at the study institution and also progressed quickly after their prior treatment (mean of 4.8 months), the argument for improved surgical expertise as the cause of the improved outcomes seems highly unlikely (Table 1). In addition, as discussed above, the literature suggests that even with excellent surgical care, resection alone frequently results in local recurrence in larger brain metastasis [1,5].
Another possibility is that the patients’ implant site LC rate was positively impacted by subsequent treatment. We believe that any impact of these treatments on the implant site control was likely small: there was only one instance of adjuvant immunotherapy and one of WBRT, and both were in the same person with the latter used when the former had failed (case 6, Table 2).
One additional shortcoming that may be raised is that we did not report the radiation biologically effective dose (BED) from this treatment. During the drafting of the protocol and again during manuscript preparation, we examined this subject in detail but ultimately concluded that BED modeling in brain tumor brachytherapy had not achieved a clear consensus. For this reason, we decided that the inclusion of a BED “number” without a full and open examination of the various current opinions and underlying mathematical models was not a wise choice and that a proper treatment of the subject was beyond the scope of this initial clinical paper. We are hopeful that in the future, brain brachytherapy BED calculations will be a useful tool, and could help place treatment outcomes in perspective with the modalities of SRS and FSRS.
We have presented our results about utilizing a novel intraoperative device designed to facilitate STaRT. In the presented patient cohort, many of whom were heavily pretreated, resection plus the use of this brachytherapy device demonstrated excellent tumor bed control with minimal side effects. No single factor seems to account for the entirety of the safety or efficacy outcomes achieved, and we believe a comprehensive explanation likely includes a) immediacy of treatment initiation, b) dose delivery at the time point of least residual tumor, c) dose accuracy, d) specified dose, and e) carrier design aspects discussed above. Given the reported outcomes, treatment utilizing this device presents a potentially effective therapeutic option for those with large and/or recurrent brain metastases, a category of tumors that currently experience relatively poor LC with existing treatment options. A commercially-produced version of the device has recently received the FDA clearance for use in both newly diagnosed and recurrent brain metastasis. To validate these outcomes, randomized clinical trials in both newly diagnosed brain metastasis (NCT 04365374, Clinicaltrials.gov) and in previously irradiated metastasis vs. standard of care treatments are opening in 2020.
- Kellner CP, D'Ambrosio AL: Surgical management of brain metastases. Neurosurg Clin N Am. 2011, 22:53-59. 10.1016/j.nec.2010.08.003
- Arvold ND, Lee EQ, Mehta MP, et al.: Updates in the management of brain metastases. Neuro Oncol. 2016, 18:1043-1065. 10.1093/neuonc/now127
- Ammirati M, Cobbs CS, Linskey ME, et al.: The role of retreatment in the management of recurrent/progressive brain metastases: a systematic review and evidence-based clinical practice guideline. J Neurooncol. 2010, 96:85-96. 10.1007/s11060-009-0055-6
- Mahajan A, Ahmed S, McAleer MF, et al.: Post-operative stereotactic radiosurgery versus observation for completely resected brain metastases: a single-centre, randomised, controlled, phase 3 trial. Lancet Oncol. 2017, 18:1040-1048. 10.1016/S1470-2045(17)30414-X
- Ling DC, Vargo JA, Wegner RE, et al.: Postoperative stereotactic radiosurgery to the resection cavity for large brain metastases: clinical outcomes, predictors of intracranial failure, and implications for optimal patient selection. Neurosurgery. 2015, 76:150-156. 10.1227/NEU.0000000000000584
- Brown PD, Ballman KV, Cerhan JH, et al.: Postoperative stereotactic radiosurgery compared with whole brain radiotherapy for resected metastatic brain disease (NCCTG N107C/CEC·3): a multicentre, randomised, controlled, phase 3 trial. Lancet Oncol. 2017, 18:1049-1060. 10.1016/S1470-2045(17)30441-2
- Gondi V, Mehta MP: Control versus cognition: the changing paradigm of adjuvant therapy for resected brain metastasis. Neuro Oncol. 2018, 20:2-3. 10.1093/neuonc/nox180
- Minniti G, Scaringi C, Paolini S, et al.: Single-fraction versus multifraction (3 × 9 Gy) stereotactic radiosurgery for large (>2 cm) brain metastases: a comparative analysis of local control and risk of radiation-induced brain necrosis. Int J Radiat Oncol Biol Phys. 2016, 95:1142-1148. 10.1016/j.ijrobp.2016.03.013
- Iorio-Morin C, Masson-Côté L, Ezahr Y, Blanchard J, Ebacher A, Mathieu D: Early Gamma Knife stereotactic radiosurgery to the tumor bed of resected brain metastasis for improved local control. J Neurosurg. 2014, 121:69-74. 10.3171/2014.7.GKS141488
- Kal HB, Struikmans H, Barten-van Rijbroek AD: Surgical stress and accelerated tumor growth. Anticancer Res. 2008, 28:1129-1132.
- Chang EL, Wefel JS, Hess KR, et al.: Neurocognition in patients with brain metastases treated with radiosurgery or radiosurgery plus whole-brain irradiation: a randomised controlled trial. Lancet Oncol. 2009, 10:1037-1044. 10.1016/S1470-2045(09)70263-3
- Blonigen BJ, Steinmetz RD, Levin L, Lamba MA, Warnick RE, Breneman JC: Irradiated volume as a predictor of brain radionecrosis after linear accelerator stereotactic radiosurgery. Int J Radiat Oncol Biol Phys. 2010, 77:996-1001. 10.1016/j.ijrobp.2009.06.006
- Choi M, Zabramski JM: Re-irradiation using brachytherapy for recurrent intracranial tumors: a systematic review and meta-analysis of the literature. Cureus. 2020, 12:e9666. 10.7759/cureus.9666
- Ware ML, Larson DA, Sneed PK, Wara WW, McDermott MW: Surgical resection and permanent brachytherapy for recurrent atypical and malignant meningioma. Neurosurgery. 2004, 54:55-63. 10.1227/01.neu.0000097199.26412.2a
- Petr MJ, McPherson CM, Breneman JC, Warnick RE: Management of newly diagnosed single brain metastasis with surgical resection and permanent I-125 seeds without upfront whole brain radiotherapy. J Neurooncol. 2009, 92:393-400. 10.1007/s11060-009-9868-6
- Shaw E, Scott C, Souhami L, Dinapoli R, Kline R, Loeffler J, Farnan N: Single dose radiosurgical treatment of recurrent previously irradiated primary brain tumors and brain metastases: final report of RTOG protocol 90-05. Int J Radiat Oncol Biol Phys. 2000, 47:291-298. 10.1016/s0360-3016(99)00507-6
- Murphy MK, Piper RK, Greenwood LR, et al.: Evaluation of the new cesium-131 seed for use in low-energy x-ray brachytherapy. Med Phys. 2004, 31:1529-1538. 10.1118/1.1755182
- Parashar B, Wernicke AG, Pavese A, et al.: Cesium-131 permanent seed brachytherapy: dosimetric evaluation and radiation exposure to surgeons, radiation oncologists, and staff. Brachytherapy. 2011, 10:508-513. 10.1016/j.brachy.2011.04.002
- Armpilia CI, Dale RG, Coles IP, Jones B, Antipas V: The determination of radiobiologically optimized half-lives for radionuclides used in permanent brachytherapy implants. Int J Rad Oncol Bio Phy. 2003, 55:378-385. 10.1016/s0360-3016(02)04208-6
- Wernicke AG, Smith AW, Taube S, et al.: Cesium-131 brachytherapy for recurrent brain metastases: durable salvage treatment for previously irradiated metastatic disease. J Neurosurg. 2017, 126:1212-1219. 10.3171/2016.3.JNS152836
- Wernicke AG, Hirschfeld CB, Smith AW, et al.: Clinical outcomes of large brain metastases treated with neurosurgical resection and intraoperative cesium-131 brachytherapy: results of a prospective trial. Int J Radiat Oncol Biol Phys. 2017, 98:1059-1068. 10.1016/j.ijrobp.2017.03.044
- Brachman DG, Youssef E, Dardis CJ, et al.: Resection and permanent intracranial brachytherapy using modular, biocompatible cesium-131 implants: results in 20 recurrent, previously irradiated meningiomas. J Neurosurg. 2018, 131:1819-1828. 10.3171/2018.7.JNS18656
- Julie DA, Lazow SP, Vanderbilt DB, et al.: A matched-pair analysis of clinical outcomes after intracavitary cesium-131 brachytherapy versus stereotactic radiosurgery for resected brain metastases (Epub ahead of print). J Neurosurg. 2020, 1-8. 10.3171/2020.3.JNS193419
- Consolidated guidance about materials licenses: program-specific guidance about medical use licenses, final report. (2019). Accessed: November 18, 2020: https://www.nrc.gov/docs/ML1925/ML19256C219.pdf.
- US Department of Health and Human Services (USDHHS): common terminology criterion for adverse events (CTCAE) Version 5.0. (2017). Accessed: June 29, 2020: https://evs.nci.nih.gov/ftp1/CTCAE/CTCAE_5.0/.
- Balan TA, Putter H: A tutorial on frailty models. Stat Methods Med Res. 2020, 29:3424-3454. 10.1177/0962280220921889
- National Comprehensive Cancer Network (NCCN): NCCN Clinical Practice Guidelines in Oncology: central nervous system cancers, version 3.2020. (2020). Accessed: October 10, 2020: https://www.nccn.org/professionals/physician_gls/pdf/cns.pdf.
- Wong JM, Panchmatia JR, Ziewacz JE, Bader AM, Dunn IF, Laws ER, Gawande AA: Patterns in neurosurgical adverse events: intracranial neoplasm surgery. Neurosurg Focus. 2012, 33:E16. 10.3171/2012.7.FOCUS12183
Resection and Surgically Targeted Radiation Therapy for the Treatment of Larger Recurrent or Newly Diagnosed Brain Metastasis: Results From a Prospective Trial
Ethics Statement and Conflict of Interest Disclosures
Human subjects: Consent was obtained by all participants in this study. St Joseph's Hospital and Medical Center IRB issued approval IRB#13RT022. This study was approved by the IRB at St. Joseph's Hospital and Medical Center. Animal subjects: All authors have confirmed that this study did not involve animal subjects or tissue. Conflicts of interest: In compliance with the ICMJE uniform disclosure form, all authors declare the following: Payment/services info: Portions of this work were supported by grants from the Arizona Commerce Authority, Phoenix, Arizona, #AZFG 2013-10 and #AZFG 2014-05, and by the Foundation for Cancer Research and Education, Gilbert, Arizona. Neither funder had any involvement in study design or in the collection, analysis, or interpretation of data. Financial relationships: Peter Nakaji, Kris Smith, Emad Youssef, Theresa Thomas, Leland Rogers, David Brachman declare(s) stock/stock options from GT Medical Technologies. GT Medical Technologies was an entity formed in 2017 to commercialize the device, and was not in existence when the trial was initiated in 2013. David Brachman declare(s) employment and stock/stock options from GT Medical Technologies. David Brachman is the Chief Technology Officer at GT Medical Technologies.
. Other relationships: All authors have declared that there are no other relationships or activities that could appear to have influenced the submitted work.
Cite this article as:
Nakaji P, Smith K, Youssef E, et al. (November 19, 2020) Resection and Surgically Targeted Radiation Therapy for the Treatment of Larger Recurrent or Newly Diagnosed Brain Metastasis: Results From a Prospective Trial. Cureus 12(11): e11570. doi:10.7759/cureus.11570
Peer review began: October 21, 2020
Peer review concluded: November 06, 2020
Published: November 19, 2020
© Copyright 2020
Nakaji et al. This is an open access article distributed under the terms of the Creative Commons Attribution License CC-BY 4.0., which permits unrestricted use, distribution, and reproduction in any medium, provided the original author and source are credited.
This is an open access article distributed under the terms of the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original author and source are credited.